Customer Support: +49 30 email@example.com
Which modules can be integrated in an assistant?
Configuration of Assistant ASM 2.1L
Is there maximum number of same moduls in an assistant?
Error message: no license available
After adding the license for my OpenLAB® CDS EZChrom Edition workstation installation I …
Which software functions does the Assistant ASM 2.1L support?
Software functions of Assistant ASM 2.1L
At which IP ports are three integrated modules connected?
Position of integrated modules and IP port connenctions
At which IP ports are two integrated modules connected?
At which IP port is one integrated module connected?
Is there a limitation in module combination?
How to change LAN settings to fixed IP address or DHCP by Mobile Control?
For most AZURA devices LAN settings can be changed by Mobile Control
How to change LAN settings to fixed IP address or DHCP by Firmware Wizard?
For selected AZURA devices LAN settings can be changed by Firmware Wizard
How to autozero a pressure offset of a P 6.1L?
Slight pressure deviations of P 6.1L can be balanced by Mobile Control.
How to upload firmware on AZURA devices?
The Firmware Wizard can be used to upload firmware on AZURA devices
How to change pump heads of pumps in Assistants?
Changing pump heads of pumps in Assistants is not automatically recognized.
How to connect the pump with the eluent?
There are connectors available for the 10 and 50 ml pump heads:3 mm ID tubing …
How to connect a flow meter to the pump?
Connect in following order:1. A analog cable from the mini Cori-Flow to the pump.2. A …
What is the maximum eluent temperature?
What is the maximum viscosity for KNAUER pumps?
The maximum viscosity of AZURA P 2.1S and P 4.1S as well as the BlueShadow Pump 40P is …
What is the optimal flowrate for the pumps?
For continuous operations we recommend a flowrate of about 40% of the maximum flowrate.
What about normal phase pump heads (organic solvents)?
There is a special version for the 10 ml pump head (AZURA P 6.1L) that is optimized for …
Which pump can I use for which solvent?
You can use all pumps under standard conditions (1 < pH < 13, not oxidative or …
What is the difference between a Dosing Pump and System Pump?
Dosing Pumps are optimized as standalone devices. Therefor they are very compact and have …
Why is the K-7400 not recognized by the EuroOsmo 7400 software?
To control the K-7400 with the EuroOsmo 7400 software the settings of the device need to …
When to use UVE photochemical derivatisation?
Enhancement of detection for aflatoxins, phenylurea pesticides, barbiturates and other …
How to deal with variable retention times?
The problem with variable retention times can occur due to following reasons:
How to deal with splitt peaks?
If you notice splitt peaks in your chromatogram, it is mostly due to column damage.